dixell. DEFAULT SETTING VALUES. تا تا تست ه تا تن تا نا م ع. XR60C signals and records temperature alarms, together with their duration and max. View and Download Emerson Dixell XR60C installing and operating instructions online. Digital controller with defrost and fans management. Dixell XR60C. Dixell Srl reserves the right to change the composition of its products, even without or faulty operation send the instrument back to the distributor or to Dixell Srl.
|Published (Last):||20 December 2009|
|PDF File Size:||10.62 Mb|
|ePub File Size:||15.68 Mb|
|Price:||Free* [*Free Regsitration Required]|
This function managed by the Fct parameter is designed to avoid short cycles of fans, that. After defrost, there is a timed fan delay allowing for drip time, set by means of the “Fnd”.
Comments to this Manuals Your Name. The fan control mode is selected by means of the “FnC” parameter:. Fit the probe where it is not accessible by the End User. Page of 4 Go.
To print the manual completely, please, download it. Two defrost modes are available through the “tdF” parameter: At the end of defrost dripping time is started, its length is set in the FSt parameter.
Universal variable speed ac drive for induction and permanent magnetmotors pages. Digital controller with defrost and fans management.
To display target set point; in programming mode it selects a parameter or confirm an. Other parameters are used to control the.
If the alarm Xrr60c is on, an alarm has taken place. To see the last temperature alarm happened; in programming mode it browses the. Ensure that the wires for probes, loads and the power supply are separated and far enough. In case of failure or faulty operation send the instrument back to the distributor or to “Dixell.
The instrument shall not be used for purposes different from those described hereunder. In case of applications in industrial environments, the use of mains filters our mod. Don’t show me this message again. Don’t have an account?
Hold dixelk SET key pressed for more than 3s, while the recorded alarm is displayed. On the display the following message is shown: Consider the maximum current which can be applied to each relay see Technical Data. Enter text from dxell To see the kind of alarm, the max min.
The instrument is fully configurable. The instrument must not be opened. In case of fault in the thermostat probe the start and stop of the compressor are timed through. Do not expose to water or moisture: By pressing ‘print’ button you will print only current page.
This manual also for: An additional parameter “FSt” provides the setting of temperature, detected by the evaporator. This is used to make sure circulation of air only.
Then the instrument displays the temperature once again. Got it, continue to print. This manual is part of the product and should be kept near the instrument for easy and quick.
Push and immediately release the SET key or wait for 5 seconds to. XR60C signals and records temperature alarms, together with their duration and max value. Push and immediately release the SET key: Unidrive hs series; unidrive m series. Add to my manuals Add.
XR60C – AS Catering Supplies
Dixell Srl reserves the right to change the composition of its products, even without notice. Check the supply voltage is correct before connecting the instrument. The instrument is provided with screw terminal block to connect cables with a cross section up nPS Pressure switch number: Flexible machine integration through communications 50 pages.
Each LED function is described in the following table. To confirm the operation, the “rSt” message starts blinking and the normal temperature.
They have three relay outputs to control compressor, fan, and defrost, which can be.